TK100 Portable Digital Multifunctional Moisture Meter For Grains, Chemical, Plastic Granule, Soap Powder, Soil, Jute, Fiber, etc.
- The Moisture Meter TK100 is a high-tech testing portable digital instrument. The newly designed moisture meter with extensive probes suits many material’s moisture content testing, just as grains, chemical raw materials, plastic granules, soap powder, soil, Chinese traditional medicine, and other fiber materials.
- The Moisture Meter TK100 utilizes the relationship between moisture content and electrical resistance to effectively measure the moisture content quickly and accurately.
Specification of TK100 Moisture Meter:
- Display:4 digital LCD
- Measuring range :0-80%
- Accuracy: ± 0.5%n
- Operation: Electrical Resistance Method, Automatic temperature Compensation
- Power supply: 4×1.5 AAA size (UM-4) battery (not including batteries)
Operation of TK100:
- Press the power button 3-10 to boot, The display shows zero, and the measurement symbol is displayed, Shown in the lower left corner of the screen; On the show is not zero, you need to school zero. When the test is zero, please let the probe hang Empty, press the zero key 3-6 to complete the zero. In the process of zero calibration, the probe can not touch. Touch any object, otherwise, zero is not allowed. Indeed School zero can be eliminated due to ambient temperature, Humidity changes, and other factors caused by the measurement error.
- Hand the instrument, the measurement probe will be vertically, Inserted into the measured object, to be digitally stable. After the show the value of the sawdust Of the moisture measurements. The probe is inserted deep, The deeper the depth (please try all the probe Insert), the more accurate the measured value, and The smaller the error.
- Press the button 3-11, left on the screen, Side appears “MAX” symbol, measured The maximum value will remain on the display On the screen if canceled, press again 3-11 key can be.
- Manually shut down and press the button 3-10 straight, To lose when the display appears OFF. Hand to shut down. 5 minutes machine When the test is in operation, it is automatically turned off Machine, to achieve energy-saving features.
- Replace the battery when the screen appears with low voltage symbols, Please replace the battery in time. Open the battery cover, according to the positive and negative direction Fitted with four batteries on the 7th.
- Press the 3-10 keys do not let go, straight
- 1 Pcs TK100
1 Pcs User Manual
Video of TK100 Portable Digital Multifunctional Moisture Meter:
There are no reviews yet.